- Like New
- Includes Carry Case
- Meet AHA 2005 Guidelines
- Fully Automatic
- CR Plus Brochure
- CR Plus AED User Manual
Simple to use and simple to own
The LIFEPAK CR® Plus AED is fully automatic, with a simple two-step process that gives the first person to respond to a SCA victim the confidence they need to help save a life. A responder only needs to:
- Turn on the device
- Apply the QUIK-PAK™ electrode pads to the victim
The fully automatic model of the LIFEPAK CR Plus AED is designed to deliver a shock only if it determines a heart needs it. No one even needs to press a shock button.
Waveform: Biphasic truncated exponential, with voltage and current duration compensation for patient impedance.*
Output Energy Sequence: Multiple levels, configurable from 150 joules to 360 joules (200 joules min for Japan). Factory default settings of 200J, 300J, 360J.
Output Energy Accuracy: ±10% into 50 ohms, ±15% into 25 to 100 ohms.
Shock Advisory System: An ECG analysis system that advises whether a shock is appropriate; meets rhythm recognition criteria specified in DF39.
The device charges for shock only when the Shock Advisory System advises defibrillation.
Typical: Thirty (30) full discharges or 210 minutes of “on time” with a fully charged device.
Minimum: Twenty (20) full discharges or 140 minutes of “on time” with a fully charged device.
Shock Charge Time: Charge times with a fully charged device: 200 joules in less than 9 seconds, 360 joules in less than 15 seconds.
System Recharge Times: Recharge times with a fully discharged device: Able to deliver 6 shocks or provide 42 minutes of operating time after 24 hours of recharge time and 20 shocks or 140 minutes of operating time after 72 hours of recharge time with a new CHARGE-PAK at temperatures above 15° C (59° F).
User Interface: The user interface includes voice prompts, audible tones and graphic prompts.
Readiness Display: The readiness display shows the device status.
OK Indicator: Shows “OK” when the last self-test was completed successfully. When the “OK” indicator is visible, all other indicators are not visible. The “OK” indicator is not displayed during device operation.
Charge-Pak Indicator: When displayed, replace the Charge-Pak™ battery charger.
Attention Indicator: When first displayed, at least six (6) discharges or 42 minutes of operating time remain.
Service Indicator: Service required when displayed.